Definitions for galvanometer constant
This page provides all possible meanings and translations of the word galvanometer constant
The Standard Electrical Dictionary
Assume a galvanometer with a very short needle and so placed with respect to its coils that the magnetic field produced by a current circulating in them is sensibly uniform in the neighborhood of the needle, with its lines of force at right angles thereto. The field is proportional to the current i, so that it may be denoted by G i. Then G is the galvanometer constant. If now the angle of deflection of the needle is ? against the earth's field H, M being the magnetic moment of the needle we have G i M cos ? = H M sin ? or i = (H/G)* tan ?. H/G is the reduction factor; variable as H varies for different places. For a tangent galvanometer the constant G is equal to 2*PI*(n/a), in which n denotes the number of turns of wire, and a denotes the radius of the circle.
Find a translation for the galvanometer constant definition in other languages:
Select another language:
Discuss these galvanometer constant definitions with the community:
Use the citation below to add this definition to your bibliography:
"galvanometer constant." Definitions.net. STANDS4 LLC, 2014. Web. 16 Sep. 2014. <http://www.definitions.net/definition/galvanometer constant>.